ATTEN ADS1102CAL MANUAL PDF
ATTEN ADSCAL vs Rigol DSe were the two options left for considering . Manual, tracking, automatic cursor measurement function;. I have recently bought an ATTEN ADSCAL+ oscilloscope. During to instructions, I get the message: “Firmware update failed, error 0xFE. Buy Now Only: US$, Buy ATTEN ADSCAL Digital Oscilloscope MHz at the ADS Series User Manual Size: ( KB).
|Published (Last):||8 February 2015|
|PDF File Size:||2.89 Mb|
|ePub File Size:||8.50 Mb|
|Price:||Free* [*Free Regsitration Required]|
When a trigger is set up properly, the oscilloscope converts unstable displays or blank screens into meaningful ads1120cal. Taking Cursor Measurements 3. The readout on the top right corner which unit is sec, show us the horizontal position. Note The oscilloscope can capture a waveform in normal YT mode at any sampling rate. You cannot adjust the horizontal position and scale of the reference waveform.
Contents in this Manual are not allowed to copy, extract and translate before being allowed by Atten Company. Screen Normal set to normal mode. ads1102csl
Featured Category Find more Fluke products,You can use side bar filter to look for attten need easier and fast. Automatically sets the oscilloscope controls to produce a usable display of the input signals. Page 82 Table Auto measure function4-Delay measure menu: Page 60 Set trigger type of the vertical channel signal to Mode Pulse Pulse trigger.
Picture Table Mask Setting function menu 1: Chapter3 Application Examples This section presents a series of application examples. This aspect has been sent to the manufacturer and they said they would work on it.
Atten ADS1000 Series Manual
Manual Mode Table Manual cursor function menu: Time Use cursor to measure time parameters. Chapter 4 prompting messages and troubleshooting 4. Mention you that the trigger Level is at a limit when you turn the Trig level knob. Record Press this button to enter the waveform record menu.
Turn off the Digital Filter function. Do not touch exposed connections and components when power is present. The cross cursor sets the position on the waveform automatically. Default setup Page Page – Appendix D: Add to Cart Qty: Trig level at limit!
To avoid electric shock when using the probe, keep fingers behind the guard Note on the probe body. Contact the site with comments or questions. Support the sampling rate: The shapes with two arrows beside them represent the front panel knobs. Page 6 General safety summary Stten Exposed Circuitry.
MATH function control button. We should have purchased a Rigol scope instead. Page mV – 5V: Do Not Operate in an Explosive Atmosphere. Software version List the software version.
Aidetek: ATTEN Mhz Oscilloscope 1G sample rate ADSCA
Table MATH function menu: Press the top second option button. By pressing ‘print’ button you will print only current page. Chapter 5 Service and Support It is also a arrow on the scale top to show the vertical position. The oscilloscope has seen a trigger and is acquiring the posttrigger data. Taking Simple Ads1102cap 3. When Select trigger condition.
When Select how to compare the Positive pulse width less than pulse trigger pulse relative to the width setting value selected in add1102cal Set Pulse Width option. Page 41 If you input a sine signal to channel 1, follow these steps: Page Picture To record waveforms, follow these steps: Save Mahual created mask setting. Same with the software. Peak Detect mode capture the maximum and minimum values of a signal Finds highest and lowest record points over many acquisitions.
Page 48 Picture Sync Line Num All lines Select appropriate video sync.
Because the measurements use the waveform record points, they are more accurate than the graticule or cursor measurements.