EPPLEY PYRHELIOMETER PDF
Tests were made of the effect of several variables on the performance of the Eppley pyrheliometer. The tests showed: (1) How the output increased with. Find out information about Eppley pyrheliometer. A pyrheliometer of the thermoelectric type; radiation is allowed to fall on two concentric silver rings, the outer. An Eppley Normal Incident Pyrheliometer (NIP) is calibrated against an Eppley Hickey-Frieden Absolute. Cavity Radiometer under a variety of environmental.
|Published (Last):||4 June 2017|
|PDF File Size:||2.96 Mb|
|ePub File Size:||10.3 Mb|
|Price:||Free* [*Free Regsitration Required]|
Normal Incidence Pyrheliometer – The Eppley Laboratory, EPLAB
You can improve Exodictionary by cleaning up the page markup and verifying that the definition is correct and then removing this tag. Canada Department of Transport Meteorological Branch Eppley is still making radiation measuring equipment, although they no longer look like this one: Retrieved from ” http: Lampblack has good absorption in both wavelength intervals, and due to its greater absorption of the solar wavelengths its equilibrium temperature on exposure to solar radiation is higher than that of the magnesium oxide.
Eppley Pyrheliometer Serial No. The similarity of the receivers with regard to absorption of long wave radiation tends to minimize the effect of any long wave radiation as from the glass cover which may fall on the receivers.
A system of thermocouples thermopile is used to measure the temperature difference between the rings. Info as printed on the plate: You can help ExoDictionary by expanding, updating, or correcting it.
This autostub has not yet had its initial categorization proof and may be categorized incorrectly.
pyrheliometrr Eppley is still making radiation measuring equipment, although they no longer look like this one: One ring is coated with magnesium oxide; to the underside of this the electrically insulated cold junctions are attached in close thermal contact.
Le contenu de ce site est sous copyleft The content of this site is under copyleft Der Inhalt dieser Website steht unter Copyleft.
This instrument has been adopted by the U. The Eppley pyrheliometer consists of a pyrheliomeger mounted under receivers inside a clear glass spherical bulb about 3 inches in diameter. This definition article has been automatically generated.
Normal Incidence Pyrheliometer
This autostub has not yet had its initial copyediting proof and may contain significant formatting and even factual errors. The receivers are concentric flat metal rings exposed in a common plane; the rings are thermally insulated from each other and from the mounting.
The magnesium oxide has a high reflectivity for radiation in the solar wavelengths and is a good absorber and emitter in the longer wavelenghts, making for a low equilibrium temperature on exposure to solar radiation. The thermocouples are platinum-rhodium 90 – 10 percent and gold palladium 60 – 40 percent and the thermopile comprises several thermocouples. Radiation is allowed to fall on two concentric silver rings, the outer covered with magnesium oxide and the inner covered with lampblack.
Canada Department of Transport Meteorological Branch. Personal tools Log in.
Interwiki Lunarpedia Marspedia Exoplatz Scientifiction. The bulb contains dry air.
Weather Bureau for station use. The other ring is coated with lampblack; the hot junctions are similarly attached to its undersurface.
The derived value of the pyrhdliometer for the instrument is 6. Attachments are provided so that measurements of direct and diffuse solar radiation may be obtained.
The overall response of the instrument to solar radiation is an e. You can improve Exodictionary by removing inappropriate categories and then removing this tag.