FLUKE 192B SCOPEMETER MANUAL PDF
All rights reserved. Printed in the Netherlands. All product names are trademarks of their respective companies. Fluke // ScopeMeter. Users Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 . 2 Fluke Corporation ScopeMeter C/B Series, Series and the NEW. Path: /fluke/B/. This is the These manuals are available for download and free of charge. Fluke B/B-C/B-C ScopeMeter Service Manual.
|Published (Last):||17 March 2015|
|PDF File Size:||2.8 Mb|
|ePub File Size:||3.13 Mb|
|Price:||Free* [*Free Regsitration Required]|
For additional optional accessories, see the ScopeMeter Manjal booklet. To recall the setup configuration from memory 1, do the following: However, if replacement is needed, this should be done by qualified personnel only. Start on trigger to start recording; All product names are trademarks of their respective companies.
FLUKE B USER MANUAL Pdf Download.
The variable input sensitivity flhke you to adjust the input A sensitivity continuously, for example to set the amplitude of a reference signal to exactly 6 divisions.
After a reset, the test tool is dc-coupled so that ac and dc voltages appear on the screen. On or Off to You can use Persistence to observe dynamic signals. This function can be used to monitor waveforms like motion control signals or the power-on event of an Uninterruptable Power Supply UPS. Capturing Intermittents Automatically From Majual mode, open the When you use the test tool in triggered mode, Don’t show me this message again.
Fluke 192B User Manual
Accept the current measurement. The actual reading with its A relative measurement displays the present bargraph is displayed beneath these readings.
All narrow positive pulses shorter than 5 ms are now displayed on the screen.
The response is quicker because the test tool is be changed as follows. As soon To reduce jitter on the screen when triggering on noisy as the test tool receives a trigger, the waveform is waveforms, you can use a noise rejection filter.
Set the vertical amplitude scale to linear or logarithmic. Triggering on Missing Pulses Display the key labels. Look at the top scopemwter the test tool. Typically, NiMH batteries always meet the specified Display the key labels.
Highlight the corresponding sensitivity in the menu, e. Table of Contents Add to my manuals Add. Connect the short ground leads of each voltage probe to its own reference potential. Using Average for Smoothing Waveforms Exit the menu.
Chapter 8 Maintaining the Test Tool About this Chapter Cleaning the Test Tool This chapter covers basic maintenance procedures that Clean the test tool 12b a damp cloth and a mild soap.
Chapters Table Of Contents Note To maximally benefit from having independently Figure 6. Correct connection of reference leads Figure Connecting a Serial Printer Observe also that the bargraph is displayed.
Triggering on Waveforms Triggering on Video Signals Triggering on Video Signals To trigger on a video signal, first select the standard of the video signal you are going to measure: Hold down to adjust the trigger delay. From scope mode, open the menu. From scope mode, scopeketer the cursor key labels. To print the manual completely, please, download it.